Very disappointed that the V2 kits don’t have the NO2 sensor (as the V1 kits did). Didn’t know until we have received the kits that they didn’t. This is probably the most useful pollutant measurement and is used by UK government and local authorities, When will you make an NO2 sensor available, at the moment this has very limited use and seems to have gone backwards since the V1 kits
Thanks for posting your opinion.
We have an explanation for this decission in this page of the documentation. TL;DR: the results from the previous metal oxyde sensors for targetted pollutants were not satisfactory, and we decided to simplify the base solution for the urban board (consumption, cost, development effort).
Now, we agree that NO2 sensors are important for urban pollution monitoring, and we won’t be left stranded without it. For that reason, a more stable solution was implemented in the Gas Pro Sensor Board, using Electrochemical Sensors from Alphasense Ltd during the iScape project. The solution implements not only NO2, but also CO and O3. The links above provide all the information about this board and the algorithm behind them. A single cell solution is also under development, similar to that ISB of Alphasense.
In terms of availability, there is no fixed date yet since the project that allows us to develop this is still going on, and we are in the final phases of the evaluation of these sensors.
Let us know if this helps.
Yes thanks. As I am sure lots of people are telling you how Nitrogen oxides are so important, so a option ASAP would be great. As the Gas pro board has already been successfully deployed can that not be made available now?
Yes indeed, we are aware of the importance of NOx, as well as the difficulty to measure them. After some iterations, we learnt that Metal Oxyde sensors are not there yet, due to their difficulty to not be affected by temperature, humidity, Ozone, or many other things.
One of the reasons that this board is not released yet is that the post-processing required for the calculation of the pollutant concentration cannot be performed inside the Smart Citizen Kit and has to be calculated in a bigger machine a posteriori, i.e. the Platform. Since this processing is still under validation, we would not like to release the hardware yet. You could say it’s half baked.
In the meantime, what I can give you is some information about the calculation process, should you be interested about it, and some reference to devices that have been deployed for quite a while in the UK, related to your question here:
Finally, another thing you could do, in case these NO2 measurements are somehow urgent and cannot wait, is to prepare a solution including an I2C ADC like this from Adafruit and the ISB from Alphasense with a 4-electrode cell for NO2. This could be connected to the auxiliary port from the SCK and log the data, as seen here. This could be supported soon in a future version of the firmware, but it’s currently not implemented.